View and Download HP A calibration manual online. Synthesizer/Function Synthesizer HP A Operating And Service Manual. Frequency ( pages) . The Operating and Service Manual contains infor¬ mation required to install, operate, test, adjust, and serv¬ ice the Hewlett-Packard Model A. Find great deals for HP a Function Generator Operating and Service Manual . Shop with confidence on eBay!.
|Published (Last):||19 April 2007|
|PDF File Size:||10.39 Mb|
|ePub File Size:||5.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
Set the digital voltmeter as follows: This message remains in effect so bit is typically mnaual by devices that devices subsequently ad- in response to a parallel poll dressed to listen will go into operation.
If the modify key is held, the digit will continue to increment or decrement after a slight delay. Frequency automatically changes to 10kHz.
HP 3325a Function Generator Operating and Service Manual
Completely define the operation s the system is required to perform. R A2C4 ?
Repeat steps d and e. Continuous sweep is start to stop, start to stop, etc.
Otherwise, the front handle kit is available separately by its -hp- part number. If A14C is changed, repeat steps d and g. AS – Past to Present.
HP 3325A Calibration Manual
Remote control of the A by means of an HP-IB system controller is accomplished through this connector. The warranty statement is located in the front of this manual.
Change A function to neg ramp and CRO trigger to pos. Repeat step d, adjusting A14C as necessary. The A does not have Trigger capability. Instructions contained in the warnings must be followed. When preceded by the blue prefix key, this key causes the A HP-IB address to be displayed in decimal code.
HP A Service Manual download, schematics, eeprom, repair info for electronics experts
Page of 23 Go. Set the A as follows: Determine the value of a 0 using the equation: Adjustments Required After Repair. This annunciator is on when the rear panel AUX output is active see Paragraph The kanual procedures compare the instrument operation to its specifications, listed in Table The A address switches are located inside have identical stacking connectors on both ends so that the top cover near the center of the instrument.
This is a basic program statement which accomplishes the following: Basic Block Diagram, Logic Circuits.
If the original packaging has been retained, pack the instrument in the same manner as it was received. Change function to pos ramp and frequency to Hz.
High speed DC voltmeter -hp- Model A. Schematic and Board Location of R11 and The output level may also be selected or displayed in V rms or in dBm 50 ohms.
HP A CALIBRATION MANUAL Pdf Download.
If the A6 board in the above instruments is replaced, the connectors on the older destination assemblies A3, A14 4A21 l will have to be changed also. Set the dc offset to the positive and negative voltages shown below. Output Amplifier Serial Numbers Servlce the delay to see the ramp reset jitter and read the positive ramp jitter nanual microseconds.
Modulation frequency may be 0 to 50 kHz. General Interface Management Lines.